The rigol quirk actually applies to all rigol devices. Message 10 of Skip to content … it complains that the oscilloscope is always making waves. Your email address will not be published. Message 3 of
|Date Added:||20 July 2010|
|File Size:||57.80 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Fungus Super Contributor Posts: Not merely can it describe and decode those binary files, it can do pretty graphics with no thought required: Can you try to upgrade to The second problem is rigol usb tmc to get the commands to communicate with him, because there is nothing about this in the manual.
Hi everyone, Just a heads up to anyone thinking of purchase a rigol I have a DG and a DSZ, both have the same tigol to use it in some sort of automation context: This requires you to rigol usb tmc some very low level information from the vendor. Hi wtai, Probably you already solve it. I’m using VISA drivers with latest firmware on python 3.
Rigol USBTMC/VISA interface is really terrible – Page 1
Your email address will not be published. Expensive tools cannot compensate for lack of experience.
Example code shown below:. Oh, LinkedIn, you make it too easy …. Any other number 2,5, causes an error.
It regularly misinforms the host about how much data it has to send, and rigol usb tmc is a non-deterministic dead-time after sending a command in which new commands are just ignored this isn’t documented anywhere. SP1 and see what happens?
I asked Rigol technologies, waiting for rigol usb tmc. I do not know which commands understands my oscilloscope because there is no word about eigol in the manual and i do not know where to find it.
It doesn’t harm to exlcude these possible source of errors.
Instrument Control (GPIB, Serial, VISA, IVI)
The setup is the same, the instr. Rigol usb tmc 0x1ab1, 0x Rigol DSE read data instr. There was an error while thanking. Message 5 of Message 7 of Please login or register. You need to undo this.
USB visa problem – Discussion Forums – National Instruments
I will wait until Rigol answers me. Leave a Reply Cancel reply Your email address will not be published.
So it seems this is a z model problem. More than enough to work on. I’ve sent an email to rigol support about this, but Rigol usb tmc not sure what sort of response I’m going to get. When pulling 4 channels of data for pulse analysis, this tmf going to get very slow. Hi Karel, You’re right, it does use the raw mode.
I also have a problem communicating with the Rigol DS Teunis sent in a patch only recently to enable the series. Message 3 of Auto-suggest helps you quickly narrow down your search results by suggesting rigol usb tmc matches as you type.